easy to use designed for both laboratory and mobile use can determine all coagulation parameters maintenance-free Starting Automatisch werden alle wichtigen Funktionen überprüft. Incubating The measuring cuvette, with the sample or reagent, is placed into one of the incubation positions. Measuring Upon completion of incubation, the cuvette is placed in the measuring channel. The measurement is then started by using either a special Start-pipette or by using the countdown timer in connection with any commercial pipette. Upon completion of the measurement, the results are automatically calculated and displayed. The mechanical measuring system A sensor monitors the ball rotating on the bottom of the cuvette. Three parameters lead to the positive detection of a developing coagulum: A large coagulum is created which stops the ball on the wall of the cuvette. A softer, slower coagulum build-up directs the ball towards the centre of the cuvette. A change in the samples viscosity causes a delay in the balls movement. The flexibility In the laboratory, at the patients house, or on the move: The Thrombostat can be used almost anywhere. It works with all standard reagents and sample types.
- Operation: : semi-automatic
- Performance: : 2-channel