SPECIFICATION: 1. Normal Working Conditions a). Surroundings temperature: 10~30 degree; b). Relative humidity: =70%; c). Atmospheric pressure: 700hPa~1060hPa; d). Voltage of electrical source: 220V±10V; e). Frequency of electrical source: 50Hz±1Hz. 2. Surroundings of transportation and reservation a). Surroundings temperature range: -40~+55 degree; b). Relative humidity range: =93% c). Atmospheric pressure range: 500hPa~1060hPa. 3.Performance (1). Processing time of film The time which develops films needed is 2min~8min, and can be accommodated continuously. The error is within±10%. (2). Light leaking: During the bathing process, light leaking is not permitted. (3) The stability of film delivery The delivery mechanism of film should not lacerate or lock films. (4) The timing panel of transit film The timing panel of transit film adopts the number 1~12 represents month, each scale represents 10 days. 4, The product structure It is composed of Main body , Light-tight box, Delivery mechanism of film, Liquid cup and Screen and so on. 5, Input power: 4VA.
- Type of support:dental radiographic films